Magnetic Flaw Detector (Magnetic Yoke) NOVOTEST MPD-DC
Magnetic Flaw Detector (Magnetic Yoke) NOVOTEST MPD-DC applies in circumstances where the electric equipment mustn’t be used or prohibited by the rules.
Device is used during magnetic particle inspection (where it is applicable) according to ASTM E 709, ASTM E 1444, ASME Section V Article 7 and MIL-STD-1949.
Magnetic Flaw Detector NOVOTEST MPD-DC is used to detect surface and subsurface cracks of all kinds (flake, lack of fusion welded joints, tears etc.) in structures made of ferromagnetic materials.
The device has two permanent magnets placed in a cylindrical shells, which are connected by a flexible magnetic wire, so it can be used for MPI of remote locations, corner welds and other products of various shapes and sizes.
Magnetic Flaw Detector NOVOTEST MPD-DC can be used in cases if user can not use electromagnets, when supply of power is difficult or restricted by rules of safety (during works at height; working in the field conditions; testing of internal surface of tanks, boilers; etc.) at operating temperature from 30 °C below zero to 50 °C above zero and relative humidity up to 95% at 35 °C .
|Magnetic field strength at the center of the air gap between the poles||200 kA / m|
|Magnetic field strength in the air in the middle region between the working ends of the magnet, not less||65 kA / m|
|Lift force, not less||22 kg (50 pounds)|
|Interpolar distance||350 mm|
|Weight of the device, not more||0.6 kg|
|Dimensions – cylindrical – flexible magnetic core||40*120*360 mm|
Operating manual of Magnetic Flaw Detector NOVOTEST MPD-DC.pdf
- Magnetic flaw detector separately
- Magnetic flaw detector; set of materials for magnetic particle inspection; magnifying glass; flashlight.
- Magnetic flaw detector; set of materials for magnetic particle inspection; magnifying glass; flashlight and control sample for the MPI.
- Magnetic Flaw Detector
- Operating manual